Determining octane numbers for motor petrols. Complies with ASTM D 2699-86, ASTM D 2700-86.
Determining cetane numbers for diesel. Complies with ASTM D 4737-03, ASTM D 613, EN ISO 5165.
Determining solidification temperature and diesel fuel type.
Content of pour-point depressants for diesel fuels.
Content of kerosine in diesel fuel.
Petrol breakdown time (oxidative stability). Complies with ASTM D 525.
New oil products grade analyzer is manufactured on the basis of advanced high precision microprocessor. the instrument can be used together with a PC a laptop through USB interface. Analyzer uses automatic computer-based calibration, saving measured data in the instrument memory with analysis date and time followed by their transfer to the PC (in Microsoft Excel or txt format). The instrument has easy of control attrition and aggressive environments resistive 8-key antiglare keyboard (made in Germany).
When determining petrol breakdown time, the petrol brand is selected in the instrument. The instrument switches to corresponding operating mode. Oxidative stability determination complies with GOST 4039-88 (ASTM D 525)
The instrument uses the timer microchip, which allows storing measurement results with analysis date and time. These data can also be transmitted to the computer.
New computer software is included into the set.
- measurement results are reflected on computer screen in convenient form;
|Parameter description||Measurement units||Value|
|Measured petrols octane numbers range||
|Acceptable limit of octane number measurement basic error, max||
|Limit of acceptable difference between parallel octane number measurements, max||ON||± 0.2|
|Petrol oxidation breakdown time measurement range||min.||50-2400|
|Acceptable basic error limit of petrol oxidation breakdown time||%||5|
|Cetane numbers measurement range||CN||20–100|
|Acceptable basic error limit of cetane numbers, max||CN||±1.0|
|Acceptable difference limit between cetane numbers parallel measurements, max||CN||± 0.5|
|Acceptable error limit when determining diesel-fuel pour point||C o||± 2|
|Kerosine content determination range in diesel fuels||%||0-95|
|Mode for pour point depressants content determination for diesel fuel||%||3|
|Acceptable basic error limit when determining kerosene content in diesel fuels||%||0.2-1|
|Acceptable basic error limit when determining of pour point depressants content||%||0.01|
|Insufficient power supply indication operation threshold||V||5.4|
|Instrument useful life||years||6|
|Overall dimensions of:|
|sensor #1 and #2:||mm||60х100|
|Instrument mass with one sensor/two sensors||gr||850|
Electronic computing module
Tool canvas bag